|Power Supply||230 V AC,50 Hz|
|Indication||4 Digits with Auto Polarity. Red Led Dispaly of 12.5mm character height.|
|Tare||Full Range Push Button Tare|
|No of Set Points||Maximum Six (Please specify no. of Set Points)|
|Control Setting||Individually settable over full range by keyboard.|
|Control Output||1 Changeover relay output. Contact rating 3 Amps,230V AC resistive load. Maximum 6 Relays.|
|Programmed Memory||Programmed stored in RAM is protected by Battery back-up.|
|Retention||Battery kept on charging when power is ON.|
|Ambient||10 to 50 Deg C|
|Size||92mm (W) X 186mm (H) panel cutout having 280mm Depth.|
|Relay Operation||Each relay will operated as per the set point. Relay Operations can be programmed with adjustable ti|
|Load Cell Excitation||10 V DC|
|Range||Up to 5000 Counts|
Indication : 5 Digits with Auto Polarity Red LED Display of 12.5mm character height. Input Calibration : LOAD CELL.
Load Cell Excitation : Provided for Load Cell
Range : Range, resolution, decimal point, list count, zero shift & calibration (single point on site) can be done by key switch with password protection.
Indicating Accuracy : 0.05% of full scale.
Resolution : 5000 / 10000 / 50000 count can set before calibration. Zero Key : Full Range Push Button provided display value tare to zero. Setting : Alarm, Trip set point can be set by key switch.
Programme Memory : Calibration & settings variable store in memory - protect at power failure. Out Put : Alarm & Trip relay 1 NO contact for 3 Amps. resistive load.
Ambient : 10 to 50°C.
Power Supply : 110 / 230VAC, 50Hz. 10% or 24 VDC.(Any one provided) Power Consumption : 5 VA.
Size : 92 x 92mm panel cut out with 150mm depth. Bazel size 96mm (w) x 96mm (h) of MS chassis with ABS cover.
Protection : Front panel meet to IP-55 & Back panel meet to IP-20.
* Up to 4 Set Points & 4 Relays contact available for control action for various applications.
* Without Display unit also provided.
* 0 to 10 VDC or 4-20mA output corresponding to specified scale provided.
* RS485 MODBUS RTU Protocol.
* External signal to be provided to start sequence.
* Adjustable time delay provided for between process sequences.